Install Environments Conditions
When the users received the chambers, they must choose an suitable using environments for the chambers. In their laboratories, rooms, plants or other areas, anywhere, the using place must meet following conditions:
1) An separated power supply for chamber, this power supply is just used with chamber.
2) Water source and drainage outlet. If the chamber running need adding water, so users should ready for the suitable water source, and ready for an drainage outlet channel.
3) Enough space. Around the chamber place must have enough distance, we suggest the distance is more than 1 meter, so it can assure the refrigeration performance and easy to maintain.
4) If the chamber need the extra devices, such like cooing tower, air compressor, vacuum pump or others, users should ready these devices in advance before using.
5) The correct environment conditions are followings:
Using environment temperature range: +5℃ ~ +35℃
Using environment relative humidity range: ≤85%rh
Atmospheric pressure: 86 ~ 106 kPa
Ground Load Bearing: ≥800kg/m2
6) Prohibited environment conditions are followings:
Strong vibration environment
Strong magnetic field environment
Flammable, explosive, corrosive and dust environments